MemberMay 7, 2020 at 9:46 am
Hi. I have a handicare xclusive stair lift and have just got a U fault code.
[U] No Float Indication V1.19 Software & Later Only Faulty Power Supply Check Power Supply
Is this a failed power supply or something else? The charging points are showing 33v. Is it the charger that switches to float or is it the PCB that regulates it? If I have 33v and it says its charging but no float, is that a new PCB? Thanks for you time and all help gratefully appreciated. Thanks
MemberMay 7, 2020 at 1:57 pm
It’s the board that dictates the float, it isn’t a charger it is a power supply
MemberMay 7, 2020 at 9:45 pm
So i need a new board?
MemberMay 7, 2020 at 9:53 pmquote Northern Mobility:
MemberMay 8, 2020 at 7:57 am
Unfortunately looks that way, I have had faulty PSU’s that have caused the U code also.
Without having parts to swap it’s difficult, might make sense to get a handicare dealer in to save expensive mistakes.
MemberMay 8, 2020 at 4:58 pm
Thank you. I have a spare power pack so will give that a try. Where at the moment is the best place to get a new pcb if i need one?
MemberMay 9, 2020 at 11:02 am
Any Handicare dealer close to your area, unless someone on here has a spare
Log in to reply.